to reading the User’s Manual. Please read the Accutrend. Plus User’s Manual carefully before using the meter for the first time. The Accutrend Plus meter is used. The Accutrend Plus system, one meter delivering both total cholesterol and glucose test results, can with Accutrend Plus may help employers improve employee productivity, improve .. 15 Cholestech LDX System User Manual. Inverness. View and Download Cobas Accutrend Plus user manual online. Accutrend Plus Blood Glucose Meter pdf manual download.
|Published (Last):||25 June 2014|
|PDF File Size:||10.40 Mb|
|ePub File Size:||2.92 Mb|
|Price:||Free* [*Free Regsitration Required]|
Performing a measurement Plausibility check after glucose measurement: Manufactured for and distributed in the U. All instructions which require you to perform an action, in addition to very important information, appear on a colored background.
The imprinted barcode contains lot-specific information which is read and stored in the instru- C Test strip bottom side Page The left column contains an illustration of the The right column states what you should do at this instrument. Page 70 Performing a measurement Alternative blood application outside the device: For more information on sample material, refer to the test strip package insert.
Memory Memory Memory The Accutrend Plus instrument has two memory areas, each of which can be used to store up to measured values together with date, time and event flags. Performing A Measurement Performing a measurement Performing a measurement After it is switched on and coded, the instrument expects you to insert the test strip.
Quality control recommendation Preparing to perform a control test Get the test strip container for the required measurement e. Performing a measurement Alternative blood application outside the device: Quality control recommendation Display of results When measurement is finished, the result is displayed. Flap open displayed during measurement The measurement chamber flap is not closed properly.
If the problem persists, please contact your local customer support and service center. Flagging Measurements Performing a measurement Flagging measurements You can add additional information to measurements, for example, to characterize a particular value with regard to special circumstances e. Inserting The Code Strip Performing a measurement Inserting the code strip Hold the code strip with thumb and index finger in the white area located at the end of the strip.
Performing a measurement Performing a measurement After it is switched on and coded, the instrument expects you to insert the test strip.
Memory Memory The Accutrend Plus instrument has two memory areas, each of which can be used to store up to measured values together with date, time and event flags.
COBAS ACCUTREND PLUS USER MANUAL Pdf Download.
Each time you press the Set button, the last stored value of each memory area, if any, is displayed. Contents Quality control recommendation Preparing to perform a control test Make sure the code strip belonging to these test strips is at hand unless the instrument has already been coded with this code strip. Cleaning If needed, you can accutrrnd the strip guide separately from the instrument under warm running water.
Set date, time and beeper. Inserting Batteries Operating the instrument Inserting batteries Quality control recommendation Performing a control test Place the instrument on a level surface or hold it in your hand. Page 29 Operating kanual instrument Ensure the instrument is switched off and turn it over. Instrument Settings Operating the instrument Instrument settings Brief overview of the instrument settings The following table provides an overview of the available settings.
Cobas Accutrend Plus User Manual
Page 75 Performing a measurement Plausibility check after glucose measurement: Press the Set button to switch to the code number for the other parameter. Display And Symbols The Accutrend Plus instrument Display and symbols Every time mnaual instrument is powered on, accutreend should check the display.
Make sure that your assignments are unique and reproducible. Performing a measurement Switching on the instrument Place the instrument on a level surface or hold it in your hand. The instrument is suitable for professional use as well as for self-testing. Alphabetical index Alphabetical index Capillary blood Performing a measurement The time needed to evaluate the sample varies depending on the test parameter.
The instrument must be powered off before you can activate the set mode. Press the M button to show the next oldest measured value within one memory area. Preparing To Perform A Measurement Performing a measurement Accutreend to perform a measurement Get the test strip container for the required measurement.
Quality control recommendation Applying the control solution 10 Apply a large free hanging drop of either the glucose or cholesterol low level control solution directly from the bottle to the test strip.
If the plhs contains no stored values, two dashes are displayed. Generally, you should first try the following solutions.
Operating the instrument Replace and close the battery compartment lid. Check the expiration date of the test strip. Blood Collection Performing a measurement Blood collection Lance the outer side of the fingertip using the lancing device to obtain a large hanging drop of blood.